View and Download ABB Advance Optima Uras 14 service manual online. Advance Optima Uras 14 Measuring Instruments pdf manual download. Manuals and User Guides for ABB Advance Optima Uras We have 1 ABB Advance Optima Uras 14 manual available for free PDF download: Service Manual. Spare Parts for ABB & Siemens Gas Analyzer & Sample Handling Systems is available. Home» Products» ABB Gas Analyzer Spare Parts» Module Uras
|Published (Last):||14 May 2008|
|PDF File Size:||10.93 Mb|
|ePub File Size:||13.95 Mb|
|Price:||Free* [*Free Regsitration Required]|
The tasks described in this chapter require special training and under certain conditions involve working on the analyzer while it is open urras powered up.
The filter cell consists of a gold-plated tube. Figure Ex module connections Gas connectors Electr. By pressing ‘print’ button you will print only current page. There are three possible emitter power settings: Solenoids Solenoids General To calibrate the analyzer module from one to three optional solenoids can be installed and then set up using the Gas module menu item. Page IR detectors, continued Supplementary help Measurement ranges from – to Start and end of the reference measurement range Autorange possible Click the “Measurement range” button to enable autoranging for the measurement range Correction functions Correction value selection at the measurement range level See below for explanation.
Hardware configuration Module hardware The Uras 14 analyzer module is equipped according to the measurement task involved. ura
These charges are the buyer’s responsibility. Change Thermal Link Change thermal link Removing the thermal To remove the thermal link, proceed as follows: Page 95 Chapter 6: The large calibration measurement range should always be calibrated initially, the small range is electronically uuras from the large measurement range.
Questions and answers about this item.
The curve is calculated at 17 predetermined set points. Delivery times may vary, especially during peak periods. Module versions stainless steel tubingcontinued Uuras Analysatormodul Uras 14 Module versions: After calibration messages 1, 2, 4 or 8 may appear. Apertures Apertures Location in module The apertures and their adjustment screws are located on the main frame between the modulator and sample cells.
Uras 14 Gas Analyzer AO ABB AO *Refurbished* | eBay
Infrared detector, continued Figure Infrared detector 1 Amplifier matching jumpers 2 Preamplifier 3 IR module circuit board plug connection 4 Membrane capacitor 5 Front and rear chambers6 Positioning pins and openings 7 Fill ports sealed gas-tight and soldered Figure Preamplifier BR 1 1x amplification The hood consists of a U-shaped plate with a heating element on each side wall.
Contents Subject See page Emitter power and amplification levels must be plausible, see Chapter 7: Page Flow detector initial calibration, continued Once the data are entered iras the analyzer module initial calibration can no longer be cancelled. Hold the cell up to a light and look uraw it.
Description of functions Chapter 2: Excessive measurement signal preparation time Indication The measurement signal changes very slowly after a concentration change.
Customs fees are normally charged by the shipping company or collected when you pick the item up. Temperature detectors, continued Component configuration Component Component name e. Testing Overview This chapter describes testing of the primary measurement and influence values on Introduction the module.
ABB Advance Optima Uras 14 Manuals
Pull the thermal link T out of the base support. Page 52 IR module circuit board, continued Figure Gasanalyse Gas Analysis Pos. The maximum permissible cxl can thus be limited. IR detectors, continued Supplementary help Measurement ranges from – to Start and end of the reference measurement range Autorange possible Click the “Measurement range” button to enable autoranging for the measurement range Correction functions Correction value selection at the measurement range level See below for explanation.
Temperature detectors, continued Temperature Temperature detector 1 is used for temperature compensation of IR components. Calibration Unit Calibration unit Location in module The calibration unit is an installable option.
International postage and import charges paid to Pitney Bowes Inc. Estimated delivery within working days Seller ships within 1 day after receiving cleared payment avb opens in a new window or tab. Description of functions Overview This chapter Step Action Turn off the analyzer power supply.
URAS 14 Gas Analyzer Ao2020 ABB 2431-0-111100000000 Ao2000 **
Pressure detector General The pressure detector pressure 1 is always installed in the analyzer module and thus is always configured. None Status messages Details Possible cause Measures Contamination at sample gas collection – Clean contaminated components and point and in lines check, if applicable, sample gas preparation temperature control – Check system shutdown procedure – Check cooler Modulator Modulator Location in module The modulator is attached to the main frame.
Contents, Fortsetzung Chapter 4: They may deviate in some cases. Final IR detector Closing disk